The U.S. Food and Drug Administration (FDA) is informing consumers, health care providers, and health care facilities that the FDA is evaluating the potential for device failures (such as leaks, breakage, and other problems) with plastic syringes manufactured in China. The FDA is collecting and analyzing data to evaluate plastic syringes made in China used for injecting fluids into, or withdrawing fluids from, the body. At this time, the issue does not include glass syringes, pre-filled syringes, or syringes used for oral or topical purposes.
The FDA received information about quality issues associated with several Chinese manufacturers of syringes. We are concerned that certain syringes manufactured in China may not provide consistent and adequate quality or performance.
Recommendations for Consumers, Health Care Providers, and Facilities
At this time, while the FDA continues its evaluation of plastic syringes made in China:
- Check the manufacturing location for syringes you use or have in your inventory by reviewing the labeling, outer packaging, or contacting your supplier or group purchasing organization.
- Consider using syringes not manufactured in China, if possible. At this time, glass syringes, pre-filled syringes, or syringes used for oral or topical purposes are not included.
- If you only have syringes manufactured in China, then continue to use them as needed until you are able to use alternative syringes and closely monitor for leaks, breakage, and other problems.
- Report any issues with syringes to the FDA.
To date, the FDA is aware of quality issues from recent syringe recalls, Medical Device Reports (MDRs), and additional complaints about syringes made at various manufacturing sites in China. Quality issues reported have included leaks, breakage, and other problems after manufacturers made changes to the syringe dimensions. These quality issues may affect the performance and safety of the syringes including their ability to deliver the correct dose of medication when used alone or with other medical devices such as infusion pumps.
The FDA is working with federal partners to further test syringes manufactured in China. The FDA plans to work with manufacturers to ensure adequate corrective actions are taken, and, as necessary, may prevent syringes made in China from entering the United States.
In addition, the FDA will continue monitoring reports of problems with syringes manufactured in China.
BD this week issued a statement regarding its syringes.
"The FDA safety notice does not apply to any BD syringes. Essentially all plastic syringes BD provides to the U.S. health care system are manufactured in the United States in Nebraska and Connecticut. BD remains committed to supporting the U.S. health care system and is ready to increase production to help supply those providers who currently purchase syringes impacted by the FDA communication," said Eric Borin, president of BD Medication Delivery Solutions.